Tube Testers and Test Equipment
Quality Test Instruments for Technicians or Hobbyists
Offer good for continental USA only. Paypal buyer must have CONFIRMED ADDRESS.
USA buyers — use PayPal BUY NOW button for easy buy.
Sencore TC162 Mighty Mite VII professional tube tester with original box and original hang tag, serviced and calibrated! – $379 free ship
Here is a real nice original Sencore TC162 Mighty Mite tube tester. Includes original box and original advertising hang tag! Also includes copies of our TubeSound remastered manual and setup chart book.
The circuit has been substantially rebuilt, including all new resistors in it. All carbon load resistors tested and replaced, where necessary, to bring into 5% tolerance. Recapped the power supply and several other new caps in the circuit board. All socket pins hand-polished and then treated with Deoxit. All switches and controls cleaned with Deoxit. A grounded line cord is installed. Finally, I calibrated this beautiful test instrument.
The TC162 Mighty Mite is popular among technicians because: (1) complete solid-state design for extreme reliability, (2) full load cathode current test method with Life Test option, (3) ultra-sensitive 100-megohm leakage test, and (4) thorough shorts test due to a full compliment of lockout switches that prevent false shorts or miss real shorts. Made in USA.
Fantastic opportunity to buy a quality test instrument, with original box, nice condition, professionally serviced and calibrated.
Triplett 3413-B Tube Tester, serviced and calibrated, $269 free ship USA
Triplett was a premier test instrument manufacturer back in the good-old-tube-days. Everything they made (that I have ever seen) was built with quality and excellent craftsmanship. In my opinion, they had the highest build quality of all of the classic emission tube testers, using a large easy-to-see top quality meter, precision resistors in key parts of the circuit, quality transformer, quality switches and controls, and built-in calibration pots (in this 3413-B model). They had a factory documented calibration procedure for emission tests. All of those features together were not found in the other well-known brands of similar tube tester.
The 3413-B is an upgraded version of their classic. It has the calibration pots and separate shorts-leakage tests (shorts @ approx 200k, leakage @ 2meg). I have calibrated both the shorts & leakage tests. Emission calibration has been performed (see pics below). All resistors checked & replaced where necessary. New 3-prong grounded line cord installed. Meter protection diodes installed. Recapped. All controls, sockets, & switches cleaned treated with DeOxit. Built-in sockets from antique 4-pin through 9-pin-miniature. QUALITY TEST INSTRUMENT.
The front panel is very nice indeed, as you see in the photos. The case has been repainted and without dents. No carry handle. Clean inside and out. Copy of instruction manual will be included. Extra tube setup supplement charts can be emailed to buyer.
A true classic tube tester that is highly recommended; it does a bang-up job at the fundamentals — shorts testing, leakage testing, and culling out the weak/bad duds. See photos below, this is a gem. Tubes shown being tested are not included.
Heathkit T-3 AF-RF Signal Tracer, beautiful condition, professionally serviced, 2 new probes – $139 free ship USA
This T-3 Heathkit signal tracer is in beautiful original cosmetic condition. I have done a lot of rebuilding inside (many new caps, etc), and replaced tubes where needed. Magic eye tube is very bright. I also built 2 new probes — one for RF (radio-freq), and one for AF (audio freq).
Outstanding piece of test equipment for servicing antique radios or audio amps. Photos below show testing a tube radio, first by signal tracing at the RF stage, then using the audio probe to signal trace at the volume control. The radio is not part of the sale, obviously. As a courtesy, I can email a PDF of the manual to the buyer.
A signal tracer is one of the most useful test instruments you can own. The uses are endless. This quality Heathkit T2 has been rebuilt and has rebuilt RF/AF probe (probe is not Heathkit, but is custom-built and very nice).
This model is probably the easiest signal tracer to use, extremely simple, no-frills.
- new capacitors (including 500v rated, brand-new electrolytics in the power supply)
- Murata X1Y2 class safety line capacitor
- new polarized line cord
- new 6K6 power output tube
- new output transformer
- all switches properly cleaned for perfect continuity, including AC power switch
- I built a probe for you using the Heathkit T2 design schematic, works great
For simplicity and because one of the transformers was replaced, the signal tracer is now hard-wired as a signal tracer, which means that the seldom-used “substitution” features are not available. Honestly, no one uses those features — such as trying to use the signal tracer as a temporary output transformer, etc. In 20+ years, I never once felt compelled to use a signal tracer in that manner. If anything, eliminating those “features” increases reliability (less chance of burning out the internal transformers) and increases simplicity of use.
Photos below show me using this signal tracer to trace signals in a typical All-American-5 tube radio. (Radio not included, obviously).
EMC 206 Tube Tester – $199 free ship USA
EMC 206 is a very rare model of tube tester, and not much documentation exists. This unit is in excellent cosmetic condition in my opinion — really nice real hardwood case. I installed new feet on both sit-down sides.
This model was marketed as a mutual conductance tester, though it is not a true mutual conductance test circuit. To give you a summary, this model uses AC voltages on the plate and screen, and AC and DC on the grid. The DC bias is variable and taken from the DC bias pot; the dc bias was rectified by the 6H6 tube inside. The line control is also rectified by the 6H6 tube. Thus, it is a unique test circuit, different from a “dynamic conductance” circuit (such as a Jackson or Eico 667) because it applies a DC potential to the grid, but not a true mutual conductance test circuit such as a Hickok or B&K (which applies pulsating DC voltages to the plate and screen). The sensitivity control is only applicable to diode-rectifier testing, or amp tube testing when using the red-green indicating scale.
As noted, the only documentation that exists is a small instruction manual, a copy of which is included. No schematics or calibration procedure are known to exist, at least not to my knowledge.
As tested, the shorts sensitivity is rather sensitive, approx 1.7 megohms. The gas test is extremely sensitive (in fact, a degree of common sense and comparison with other known-good tubes will be necessary for the gas test.) Photos show test results from 5U4GB and 6L6G using the good-bad scale. Test results for these tubes are more than reasonable, as both tubes are well-known to me.
Limited servicing was done. I replaced a number of out-of-spec resistors and cleaned all of the controls. The unit is not “calibrated” because no calibration documentation exists, and no representation regarding performance is being made.
This tester has a basic roll chart, so it would be best considered as a tester of “antique tubes.” It would be a good affordable compliment to a newer-style tester — such as Mighty Mite, B&K, or similar — because you would be able to test the popular antique tubes.
Signal Corp I-177-B tube tester, manufactured by Simpson, with rare MX-949A/U socket adapter kit, nicest I-177B and MX-949A that I have ever seen or owned, tester serviced & calibrated, $799 free ship USA.
(This combo sells as a pair; I will not sell the MX-949A/U separately.)
Here is the nicest I-177B and MX-949A/U that I have ever encountered (either myself or sent to me for repair by a customer), this combo has been a part of my personal collection for close to 15 years, tester made by Simpson. The Simpson-manufactured units are extremely high quality and well laid-out inside. I am including the very rare MX-949A/U socket adapter kit. Both items are in really outstanding condition.
I have serviced & calibrated the Signal Corp I-177-B mutual conductance tube tester. Servicing consisted of replacing the #83 tube with my solid-state fused upgrade, checking the 5Y3GT tube for excellent balance with my oscilloscope, cleaning all pots, switches, controls with Deoxit. Socket pins cleaned and shine very close to when new. Many original carbon comp resistors were replaced with fresh modern carbon-film or metal-film 5% or better. Recapped. Brand new grounded line cord installed and with new rubber eyelet grommet. Line calibration performed. Calibration voltages checked & set for accuracy (150v plate 1000 ohms/v, 130v screen 1000 ohms/v, accurate bias voltage checked at several positions of the bias pot, signal voltage 5.0 vac), overall calibration checked against my calibration tubes. Diode/Rectifier/0Z4 test functions checked for accuracy. Gas test checked for accuracy. Shorts test checked for proper working condition. General cleaning inside & out.
Includes original tube data setup book. No other paperwork included, but tester instruction manual and additional supplements can be downloaded or forwarded via email PDF.
The I-177 series is a quality mutual conductance tube tester, Hickok-style circuit, and designed for testing older tubes, pre 9-pin-miniature. The MX-949A socket kit adds a lot of flexibility to testing (such as testing 9-pin-miniature and septar base tubes), but honestly it is somewhat of a pain-in-the-butt to use the socket adapter and therefore should be considered as an extra that you would use on a very rare occasion. It does come in handy if you want to test rare transmitting septar-base tubes on the rare occasion that you need to. Overall, I highly recommend the I-177 series as being great for antique radio servicing or tube collectors that require setup data for oddball old military tubes.
When considering to buy an I-177 tester, even though the socket adapter kit is included, I can tell you that the I-177 series is not ideal for someone who only wants to own a single tube tester, because testing 9-pin-miniature or newer tubes is slow and cumbersome because you need to use the MX-949A socket adapter kit. In a nutshell — if you want a top quality vintage military tester for testing primarily the old basing tubes, then this is an awesome choice. It does a great job of testing the old 4-pin, 5-pin, 6-pin, 7-pin-large, etc… types of tubes because you do not need to use the socket adapter kit. Therefore, if you need to test 9-pin-miniature tubes more than rarely, you really should look for a different tester.
What many pro’s do — they will pair this I-177B + MX949A combo with a newer tester, such as a Sencore Mighty Mite or B&K, which provides them the capability to test the overwhelming majority of USA tubes ever made.
Photos below show the outstanding condition of the I-177B tester and the MX-949A/U socket adapter kit. Photo also shows accurate testing of my 6L6 calibration tube (not included).
powerful 45-AMP Filament transformer 6v – $39 free ship USA
Here is a whopper for your tube projects! A massive 45-amp filament transformer. Nominal input 115vac, nominal output 6.5vac @ 45A. Want to build a massive tube warmer/heater? Now is your chance. With this transformer, you can power the filaments of almost *50* (yes, fifty…) 6L6 tubes at the same time! Size: approx 4L x 3.25W x 4.5H inches.